Over deze norm
1.1 These test methods cover the determination of fluoride ion in water. The following two test methods are given:
Test Method A—Distillation
Test Method B—Ion Selective Electrode
1.2 Test Method A covers the accurate measurement of total fluoride in water through isolation of the fluoride by distillation and subsequent measurement in the distillate by use of the ion selective electrode (ISE) method. The procedure covers the range from 0.1 to 2.6 mg/L of fluoride.
1.3 Test Method B covers the accurate measurement of simple fluoride ion in water by means of an ion selective electrode. With this test method, distillation is eliminated because the electrode is not affected by the interferences common to colorimetric procedures. Concentrations of fluoride from 0.1 to 1000 mg/L may be measured.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For a specific precautionary statement, see .
1.6 Former Test Method A, SPADNS Photometric Procedure, was discontinued. Refer to for historical information.
|Nederlandse titel||Standard Test Methods for Fluoride Ion in Water|
|Engelse titel||Standard Test Methods for Fluoride Ion in Water|