Over deze norm
1.1 This guide covers procedures for using the ASTM Standard Reference Blacks (SRBs) and the HT and INR Iodine Number Standards to continuously monitor the precision of those carbon black test methods for which reference values have been established. It also offers guidelines for troubleshooting various test methods.
1.2 This guide defines the environmental conditions that are required for laboratories that perform carbon black testing activities for those test methods under D24’s jurisdiction.
1.3 This guide establishes procedures for the use of x-charts to continuously monitor those tests listed in Section for within-lab precision (repeatability) and between-lab accuracy (reproducibility).
1.4 This guide provides a statistical procedure for improving test reproducibility when a laboratory cannot physically calibrate its apparatus to obtain the reference values of the ASTM reference blacks, within the ranges given in this guide.
|Engelse titel||Standard Guide for Carbon Black—Validation of Test Method Precision and Bias|