Over deze norm
1.1 These practices describe means for calibrating ambient, workplace or indoor ozone monitors, using transfer standards.
1.2 These practices describe five types of transfer standards:
Practice A—Analytical instruments,
Practice B—Boric acid potassium iodide (BAKI) manual analytical procedure,
Practice C—Gas phase titration with excess nitric oxide,
Practice D—Gas phase titration with excess ozone, and
Practice E—Ozone generator device.
1.3 These practices describe procedures to establish the authority of transfer standards: qualification, certification, and periodic recertification.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use. See Section for specific precautionary statements.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|Engelse titel||Standard Practices for Calibration of Ozone Monitors Using Transfer Standards|