Over deze norm
1.1 This test method covers the measurement of radon in drinking water in concentrations above 2 Bq/L.
1.2 This test method may be used for absolute measurements by calibrating with a 226Ra standard or for relative measurements by comparing the measurements made with each other.
1.3 This test method is used successfully with drinking water samples and Type III reagent water conforming to Specification . It is the user’s responsibility to ensure the validity of this test method for waters of untested matrices.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|Nederlandse titel||Standard Test Method for Radon in Drinking Water|
|Engelse titel||Standard Test Method for Radon in Drinking Water|