Over deze norm
1.1 This practice covers the testing of the James Machine for repeatability of static coefficient of friction, relative to a standard reference interface consisting of the working surfaces of Borco2 board and standard leather shoe sole material, or a control polish film and standard leather shoe material. The practice provides basis data on the stability of the James Machine to ensure accurate static coefficient of friction determinations over time and repeated use and for determining if the James Machine is mechanically calibrated and properly aligned.
1.2 This practice is written specifically for James Machines with manual or motorized test table transport. Variations of this practice for the calibration of versions of James Machines which are semi-automated are obvious. Calibration practices suggested by the manufacturer of semi-automatic James Machines should be followed in preference to this practice.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|Nederlandse titel||Standard Practice for Calibration of the James Static Coefficient of Friction Machine|
|Engelse titel||Standard Practice for Calibration of the James Static Coefficient of Friction Machine|