Over deze norm
1.1 This guide provides procedures for using a reference material with a known emission rate of a volatile organic compound (VOC) to estimate the bias associated with a VOC emission chamber test.
1.2 This guide may be used to assess measurements of VOC emissions conducted in a variety of environmental chambers, such as small-scale chambers, full-scale chambers, emission cells, and micro-scale chambers.
1.3 This guide may be used to assess measurements of VOC emissions from a variety of sources including “dry” materials (for example, carpet, floor tile and particleboard) and “wet” materials (for example, paint and cleaning products).
1.4 This guide can be used to support quality control efforts by emissions testing laboratories, third party accreditation of testing laboratories participating in emissions testing programs, and quality control efforts by manufacturers of building and other materials.
1.5 This guide may be used to support the determination of precision and bias of other commonly used VOC emission standards including Guide , Test Method , ISO 16000-9, ANSI/BIFMA M7.1, and CDPH/EHLB/Standard Method V1.1.
1.6 This guide also describes the attributes of a suitable emission reference material and the different methods available to independently determine the reference material’s VOC emission rate.
1.7 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.8 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|Engelse titel||Standard Guide for Using Reference Material to Characterize Measurement Bias Associated with Volatile Organic Compound Emission Chamber Test|