Over deze norm
1.1 This test method describes a means for measuring the kinematic viscosity of transparent and opaque liquids such as new and in-service lubricating oils using a miniature microchannel viscometer at 40 °C in the range of 12.9 mm2/s to 174 mm2/s
1.2 The precision has only been determined for those materials and viscosity ranges, as indicated in Section on Precision and Bias.
1.3 This test method is specifically tailored to obtaining a rapid, direct, temperature- stabilized measure of the kinematic viscosity of new and in-service lubricants in the field in real- time without the use of solvents or chemical cleaning agents. The measurement takes place at 40 °C and kinematic viscosity is directly obtained. No temperature extrapolations or density corrections are necessary.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Some specific hazards statements are given in Section on Hazards.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|Nederlandse titel||Standard Test Method for Field Determination of Kinematic Viscosity Using a Microchannel Viscometer|
|Engelse titel||Standard Test Method for Field Determination of Kinematic Viscosity Using a Microchannel Viscometer|