Over deze norm
1.1 This guide covers methods for users to test metal sheathed thermocouple assemblies, including the extension wires just prior to and after installation or some period of service.
1.2 The tests are intended to ensure that the thermocouple assemblies have not been damaged during storage or installation, to ensure that the extension wires have been attached to connectors and terminals with the correct polarity, and to provide benchmark data for later reference when testing to assess possible damage of the thermocouple assembly after operation. Some of these tests may not be appropriate for thermocouples that have been exposed to temperatures higher than the recommended limits for the particular type.
1.3 The tests described herein include methods to measure the following characteristics of installed sheathed thermocouple assemblies and to provide benchmark data for determining if the thermocouple assembly has been subsequently damaged in operation:
1.3.1 Loop Resistance:
22.214.171.124 Combined extension wires and thermoelements.
1.3.2 Insulation Resistance:
126.96.36.199 Insulation, thermocouple assembly,
188.8.131.52 Insulation, thermocouple assembly and extension wires.
1.3.3 Seebeck Voltage:
184.108.40.206 Combined extension wires and thermocouple assembly.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|Nederlandse titel||Standard Guide for Testing Sheathed Thermocouples, Thermocouples Assemblies, and Connecting Wires Prior to, and After Installation or Service|
|Engelse titel||Standard Guide for Testing Sheathed Thermocouples, Thermocouples Assemblies, and Connecting Wires Prior to, and After Installation or Service|