Over deze norm
1.1 This practice () describes three methods for determining the relative spectral correction factors for grating-based fluorescence spectrometers in the ultraviolet-visible spectral range. These methods are intended for instruments with a 0°/90° transmitting sample geometry. Each method uses different types of transfer standards, including 1) a calibrated light source (CS), 2) a calibrated detector (CD) and a calibrated diffuse reflector (CR), and 3) certified reference materials (CRMs). The wavelength region covered by the different methods ranges from 250 to 830 nm with some methods having a broader range than others. Extending these methods to the near infrared (NIR) beyond 830 nm will be discussed briefly, where appropriate. These methods were designed for scanning fluorescence spectrometers with a single channel detector, but can also be used with a multichannel detector, such as a diode array or a CCD.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
|Engelse titel||Standard Practice for Determining Relative Spectral Correction Factors for Emission Signal of Fluorescence Spectrometers|