Over deze norm
This clause of Part 1 is applicable except as follows: 1.1.1 Equipment included in scope Replacement: Replace the existing text with the following: This part of IEC 61010 specifies safety requirements for METERS. The METERS that have a primary purpose of measuring voltage on a live MAINS CIRCUIT are within the scope of this standard. They have various names, but all of them have capability for measurements of voltages on a live MAINS CIRCUIT. Some of the names given to this equipment are as follows: - MULTIMETER; - digital MULTIMETER; - VOLTMETER; - clamp METER (see also Part 2-032). For the purpose of this standard, the term METER is used for these HAND-HELD measuring instruments. NOTE Parts of the equipment that are not within the scope of this Part 2-033 are considered to be covered by the requirements of Part 1 or other part 2's of IEC 61010 and then will also need to meet the requirements of these other parts. 1.1.2 Equipment excluded from scope Addition: Add the following new item to the list: aa) IEC 61557 (Electrical safety in low voltage distribution systems up to 1 000 V a.c. and 1 500 V d.c. - Equipment for testing, measuring or monitoring of protective measures - Parts 1 through 12). Addition: Add the two following paragraphs at the end of the subclause: Equipment that is not capable of measuring MAINS voltages is not within the scope of this Part 2-033. See IEC 61010-2-030 for requirements pertaining to such equipment. 61010-2-033 Ó IEC:2012 - 7 - Such equipment, including other HAND-HELD equipment such as oscilloscopes, wattmeters, process control MULTIMETERS, and communications test sets is not within the scope of this Part 2-033. 1.2.1 Aspects included in scope Addition: Add the following paragraph at the end of the subclause: Requirements for protection against HAZARDS resulting from NORMAL USE and REASONABLY FORESEEABLE MISUSE of measuring circuits are given in Clause 101.
|Nederlandse titel||Veiligheidseisen voor elektrisch materieel voor meet- en regeltechniek en laboratoriumgebruik - Deel 2-033: Bijzondere eisen voor in de hand vast te houden multimeters en andere meters, voor huishoudelijk en professioneel gebruik, geschikt voor het meten van netspanning|
|Engelse titel||Safety requirements for electrical equipment for measurement, control, and laboratory use - Part 2-033: Particular requirements for hand-held multimeters and other meters, for domestic and professional use, capable of measuring mains voltage||Normatieve verwijzingen||