Over deze norm
|Commissie||Anesthesie- en beademingssystemen|
NEN-EN-ISO 9170-1 is intended especially to ensure the gas-specific assembly, mechanical resistance, flow, leakage and pressure drop of terminal units and to prevent their interchange between different gases and services and applies to terminal units: a) intended for use in medical gas pipeline systems in accordance with ISO 7396-1; b) used as pressure outlets on pressure regulators in accordance with ISO 10524-1; c) used as pressure outlets on pressure regulators integrated with cylinder valves (VIPR) in accordance with ISO 10524-3. This document applies to terminal units for use with the following gases for administration to patients or for medical uses (A): - oxygen (A); - nitrous oxide (A); - medical air (A); - carbon dioxide (A); - oxygen/nitrous oxide mixture (A); - helium/oxygen mixtures (A); - oxygen 93 (A);- gases and gas mixtures classified as medical device (A); - gases delivered to medical devices or intended for medical purposes (A); - gases and gas mixtures for medicinal use not specified above (A). This document applies to terminal units for use with the following gases (B): - air for driving surgical tools (B);- nitrogen for driving surgical tools (B). This document applies to terminal units for use with vacuum systems (C). This document specifies requirements for terminal units for supply and disposal of nitrogen and air for driving surgical tools.
|Nederlandse titel||Afnamepunten voor pijpleidingsystemen voor medische gassen - Deel 1: Afnamepunten voor medische gassen onder druk en vacuÃ¼m|
|Engelse titel||Terminal units for medical gas pipeline systems - Part 1: Terminal units for use with compressed medical gases and vacuum|